TY - JOUR

T1 - Mechanical model analysis for resonance shear measurement

AU - Mizukami, Masashi

AU - Hemette, Sylvain Maurice Paul

AU - Kurihara, Kazue

N1 - Funding Information:
This study was supported by the “Research and Development of Next-Generation Filed” from the Ministry of Education, Culture, Sports, Science, and Technology, Japan, and Accelerated Innovation Research Initiative Turning Top Science and Ideas into High-Impact Values (ACCEL) of the Japan Science and Technology Agency (JST). S.H. acknowledges the fellowship from Michelin.
Publisher Copyright:
© 2019 Author(s).

PY - 2019/5/1

Y1 - 2019/5/1

N2 - Resonance shear measurement (RSM), which we developed based on a surface force apparatus, can investigate the structuring, and rheological and tribological properties of confined liquids as a function of the surface separation distance (D) from several μm to zero with a nanometer resolution. Using RSM, we reported that the nanoconfined liquids, including commercial lubricants, exhibited properties quite different from the bulk phase. Mechanical modeling of the resonance system is necessary to quantitatively evaluate the properties of confined liquids and/or sheared interfaces. In this study, we improved the model for RSM as follows: (1) We directly measured the movements of the upper and lower surfaces in addition to the measurement on the movement of the vertical spring to confirm the model, which could be used to estimate the parameters used in the model; (2) we proposed a modified mechanical model which considers the effect of additional motion. This model could fit the resonance curves using a identical apparatus constant for the entire measurement range, while the characteristic values of the confined liquids are the same as those obtained using our previous model, and (3) we calculated the friction (shear) force using the improved model and obtained the characteristic values (viscous and elastic parameters) of the confined liquids. This study afforded the simplicity and reliability of a mechanical model analysis of resonance curves, and the friction force calculation employing the viscous and elastic terms can be useful for studying the mechanism of friction force.

AB - Resonance shear measurement (RSM), which we developed based on a surface force apparatus, can investigate the structuring, and rheological and tribological properties of confined liquids as a function of the surface separation distance (D) from several μm to zero with a nanometer resolution. Using RSM, we reported that the nanoconfined liquids, including commercial lubricants, exhibited properties quite different from the bulk phase. Mechanical modeling of the resonance system is necessary to quantitatively evaluate the properties of confined liquids and/or sheared interfaces. In this study, we improved the model for RSM as follows: (1) We directly measured the movements of the upper and lower surfaces in addition to the measurement on the movement of the vertical spring to confirm the model, which could be used to estimate the parameters used in the model; (2) we proposed a modified mechanical model which considers the effect of additional motion. This model could fit the resonance curves using a identical apparatus constant for the entire measurement range, while the characteristic values of the confined liquids are the same as those obtained using our previous model, and (3) we calculated the friction (shear) force using the improved model and obtained the characteristic values (viscous and elastic parameters) of the confined liquids. This study afforded the simplicity and reliability of a mechanical model analysis of resonance curves, and the friction force calculation employing the viscous and elastic terms can be useful for studying the mechanism of friction force.

UR - http://www.scopus.com/inward/record.url?scp=85066131034&partnerID=8YFLogxK

UR - http://www.scopus.com/inward/citedby.url?scp=85066131034&partnerID=8YFLogxK

U2 - 10.1063/1.5084117

DO - 10.1063/1.5084117

M3 - Article

C2 - 31153292

AN - SCOPUS:85066131034

VL - 90

JO - Review of Scientific Instruments

JF - Review of Scientific Instruments

SN - 0034-6748

IS - 5

M1 - 055110

ER -