A new position-sensitive γ-ray detector to be used in a two-dimensional angular correlation of positron annihilation radiation (2D-ACAR) apparatus has been developed. It consists of 36 compact position-sensitive photomultiplier tubes (PS-PMT: HAMAMATSU R5900-00-C8), a light guide, and 2676 Bi4Ge3O12 (BGO) scintillator pieces of size 2.6 mm × 2.6 mm × 18 mm. A high detection efficiency for 511 keV γ-ray is achieved with the length of BGO scintillators used. The detection area is about 160 mm × 160 mm. The 288 anode outputs of the PS-PMTs are wired and connected to resistor chains from which 16 outputs (8 outputs each along the X and Y directions) are taken to identify the incident position of the γ-ray. The spatial resolution is about 3 mm (FWHM). The timing signal taken from the last dynodes of the PS-PMTs gives a timing resolution of 7.7 ns (FWHM) for 511 keV positron annihilation γ-rays.
|ジャーナル||Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment|
|出版ステータス||Published - 2002 7 21|
ASJC Scopus subject areas
- Nuclear and High Energy Physics